Measurement lines |
Single-phase/ two-wires |

Measurement items | Voltage, Current, Active power, Apparent power, Power factor, Frequency, Integration (current, active power), Waveform peak (voltage and current) |

Measurement ranges | [Voltage] AC/DC 15.000/30.00/150.00/300.0V[Current] AC/DC 100.00/300.0 mA, 1.0000/3.000/10.000/30.00A[Power] 1.5000 W〜9.000 kW (combination of voltage and current ranges) |

Integration measurement Integration time up to 10,000 hours |
[Current] No.of displayed digits: 6 digits (from 0.00000mAh, Polarity-independent integration and Sum value)[Active power] No.of displayed digits: 6 digits (from 0.00000mWh, Polarity-independent integration and Sum value) |

Input resistance | 2.4MΩ for voltage, 10mΩ or better for current, (50/60 Hz, direct input) |

Basic accuracy | ±0.1%rdg. ±0.2%f.s. (DC), ±0.1%rdg. ±0.1%f.s. (45Hz to 66Hz) Note: Provided accuracy of 1 Year, typical value |

Display refresh rate | 5 times per second |

Frequency characteristics | DC, 45Hz to 5kHz |

Waveform output | Parameter output representation: voltage, current and active power (3 simultaneous channels), Output voltage: 1V DC f.s. |

Analog output (D/A output) |
Parameter output representation: voltage, current active power and selected 1 item (4 simultaneous channels), Selected 1 item from apparent power, power factor, current integration, active power integration, Output voltage: ±2V DC f.s. |

Functions | Rectification method switchable between AC+DC (True RMS), DC (simple average), AC (True RMS), Wave peak measurement, VT or CT ratio settings, Average function |

Interface | RS-232C included as standard, GP-IB (Model 3334-01 only) |

Power supply | 100V to 240V AC, 50/60Hz, 20VA max. |

Dimensions and mass | 210 (8.27in) W × 100 (3.94in) H × 245 (9.65in) D mm, 2.5kg (88.2oz) |

Supplied accessories | Instruction Manual ×1, Power cord ×1 |

Details

Measurement lines | Single-phase/ two-wires |

Measurement items | Voltage, Current, Active power, Apparent power, Power factor, Frequency, Integration (current, active power), Waveform peak (voltage and current) |

Measurement ranges | [Voltage] AC/DC 15.000/30.00/150.00/300.0V[Current] AC/DC 100.00/300.0 mA, 1.0000/3.000/10.000/30.00A[Power] 1.5000 W〜9.000 kW (combination of voltage and current ranges) |

Integration measurement Integration time up to 10,000 hours |
[Current] No.of displayed digits: 6 digits (from 0.00000mAh, Polarity-independent integration and Sum value)[Active power] No.of displayed digits: 6 digits (from 0.00000mWh, Polarity-independent integration and Sum value) |

Input resistance | 2.4MΩ for voltage, 10mΩ or better for current, (50/60 Hz, direct input) |

Basic accuracy | ±0.1%rdg. ±0.2%f.s. (DC), ±0.1%rdg. ±0.1%f.s. (45Hz to 66Hz) Note: Provided accuracy of 1 Year, typical value |

Display refresh rate | 5 times per second |

Frequency characteristics | DC, 45Hz to 5kHz |

Waveform output | Parameter output representation: voltage, current and active power (3 simultaneous channels), Output voltage: 1V DC f.s. |

Analog output (D/A output) |
Parameter output representation: voltage, current active power and selected 1 item (4 simultaneous channels), Selected 1 item from apparent power, power factor, current integration, active power integration, Output voltage: ±2V DC f.s. |

Functions | Rectification method switchable between AC+DC (True RMS), DC (simple average), AC (True RMS), Wave peak measurement, VT or CT ratio settings, Average function |

Interface | RS-232C included as standard, GP-IB (Model 3334-01 only) |

Power supply | 100V to 240V AC, 50/60Hz, 20VA max. |

Dimensions and mass | 210 (8.27in) W × 100 (3.94in) H × 245 (9.65in) D mm, 2.5kg (88.2oz) |

Supplied accessories | Instruction Manual ×1, Power cord ×1 |

- 质量承诺
- 正品保修
- 送货到家
- 交易简单化